Experimental Control System Home Page
Current Hardware Settings
The Biopac GSR amplifier has several switches. The switches are typically set to the following values, which should be reasonable for general usage. Hardware switches should not be changed by individual users.
Gain = 5 micromho/V … for expected DC response between 0-50 micromho
Low Pass = 1 Hz … sufficient for almost all GSR studies
High Pass = Direct Current … no filtering provides absolute readings
Software interface for recording GSR